PRECISION TESTING INSTRUMENT Phone:+86-769-89201068 E-mail:firstname.lastname@example.org
Salt spray test why obviously enough water to show low water level
There are three possibilities:
1, the new salt water, from the medicine box to the spray box has a time, warning lights will be extinguished after a while.
2, plus medicine box to the spray box pipe blockage, if so, spray box without salt water. Remove the spray tower to see if there is any water below.
3, the water level sensor is broken. Only to find salt spray test machine manufacturers repair.
CONTACT PERSON : Ms.Zhong
TEL : +86-769-89201068
PHONE : +86 13829267288
OFFICE ADD: Caibai No.1 Industrial Zone, Daojiao Town, Dongguan City, Guangdong Province, China
LEAVE A MESSAGE